Integrated Processor Control:PC 104 based processor, Graphical User Interface – 38.1 cm (15 in) color touch screen
Dimensions:54 cm x 59 cm x 132 cm (21.1 in x 23.2 in x 51.8 in) (to top of monitor)
Space Requirements:When in use, aprox. 20 cm (7.9 in) of space should be left clear around unit to allow adequate ventialtion.
Shipping Weight:55.3 kg (122 lbs)
Console Weight:24 kg (53 lbs)
Custom Cart:22.2 kg (49 lbs)
Electrical:100-120V~, 450V A, 50/60Hz, 200-240V~, 450V A, 50/60hz
Operating Environment:Humidity – Non-condensing humidty below 75% RH; Temperature – 10 degrees C to 29.4 degrees C (50 degrees F to 85 degrees F)
XV Handpiece LightOutput:
Wavelength:915 nm (Laser Radiation); 650 nm (LED Light)
Maximum Radiant Power:15 W @ 915 nm; 1 W @ 650 nm
Laser Aperture:70 mm x 40 mm
Beam Profile:Non-collimated, highly diverging, diffuse beam
Power Density:0.57 w/cm2
XV Handpiece Massage:
Vacuum Level:Variable from 34mbar to 237mbar; Variable from 1 in/hg to 7 in HG
Flow Rate:18.5 m3/hr (653 ft3/hr)
Roller Diameter:32 mm (1.26″)
Roller Centers:72 mm (2.83 in)
Petite Handpiece Light Output (Same as XV, Except For):
Maximum Radiant Power:10W +/- 20% @ 915nm, 1W +/- 20% @ 650nm
Laser Aperture:45mm x 28 mm (1.77″ x 1.10″)
Petite Massage (Same as XV, Except For):
Roller Diameter:19mm (0.75″)
Roller Centers:55mm (2.16 in)
SmoothShapes 915nm laser penetrates into tissue and is preferentially absorbed by lipids1causing a thermal effect. The devices vacuum component positions the skin for optimal laser and light penetration.
650 light has been demonstrated to affect permeability of the adipocyte (fat) cell membranes2, allowing treated fat to move into the interstitial space, evacuating the expressed fat and fluids into the lymphatic system.
A 915 nm laser penetrates into tissue and is preferentially absorbed by lipids, causing a thermal effect. The temperature inside the adipocyte is slightly elevated.
915 nm laser is one of the four wavelengths identified to be absorbed by lipids at a higher rate than water.
650 nm light modifies the permeability of fat cell membranes, allowing treated fat to move into the interstitial space. This change does not damage the adipocytes.
SmoothShapes XV has created a scientific approach to cellulite treatments that practitioners and patients can trust for results. The SmoothShapes XV System effectively treats cellulite through proprietary technology called Photomology. Photomology restores enlarged cells through a unique mechanism of action that combines light and laser energy with mechanical manipulation (vacuum and massage) to specifically target problem cellulite. Treatment results: tighter, smoother looking skin.
The lightweight, handheld roller glides over the treated area, creating a gentle massage-like sensation.
A vacuum component positions the skin more effectively for optimal light and laser penetration.
The contoured rollers assist in moving the evacuated fat and fluids from the interstitial space into the lymphatic system.
A combination of light and laser energy penetrates into the tissue causing a thermal effect and increasing cell permeability.
Fat molecules migrate from the adipose cells to the interstitial fluid.
SmoothShapes PETITE Laser Handpiece Features:
Lightweight, for use with SmoothShapes XV System
Approximately 50% increase in power density to produce an identical thermal gradient as the larger laser handpiece ensuring same clinical efficacy
Improved suction adjustment